|Property||Nominal Value||Specified Range|
CB1: 870 - 930
CB2: 870 - 930
|Mean Width /µm||
CB1: 2x13 - 2x15
CB2: 2x27 - 2x29
CB1: 95 - 105
CB2: 195 - 205
|Force Constant /(N/m)||
CB1: 0.15 - 0.45
CB2: 0.06 - 0.12
|Resonance Frequency /kHz||
CB1: 42 - 58
CB2: 16 - 24
|Order Code||Number of AFM probes per pack||Data sheet||Coating|
|qp-BioT-10||10||without||partial Au coating|
|qp-BioT-20||20||without||partial Au coating|
|qp-BioT-50||50||without||partial Au coating|
Due to their unique geometry the tips of the uniqprobes are more susceptible to tip damage by electrostatic discharge (ESD) than other Silicon-SPM-Probes.
Electric fields near the probe chip may lead to field evaporation which can blunt the tip apex of the probe tip. Therefore the NANOSENSORS™ uniqprobes are shipped in specially designed ESD-safe chip carriers.
NANOSENSORS™ recommends to their customers to take appropriate precautions to avoid tip damage due to electrostatic discharge when handling the probes. This can for example be done by using anti-electrostatic mats, wrist bands and tweezers.